Over the last several years, Biosafety Cabinet (BSC) manufacturers have lowered downflow velocity while maintaining inflow velocity to increase energy efficiency, reduce exhaust system requirements and lower BSC Noise and vibration levels. However, the lowering of downflow velocity may come with a consequence of greater lateral airflow movement within the work zone creating more significant potential for cross-contamination. The lowering of the downflow velocity does not compromise personnel and product containment performance. Why? Primarily, by maintaining inflow velocity above a certain level as required by the standard, typically 100 fpm (.51 mps). However, the standard does not require a specific downflow velocity and can be adjusted as needed based on the design parameters of the work zone supply diffuser and grill patterns (airflow distribution) to assure compliance to the personnel and product protection requirements of NSF/ANSI 49. NSF/ANSI 49 test procedures do test for cross-contamination, but only at the work zone sidewalls, which doesn’t necessarily translate into the lateral movement of airflow in the center of the work zone. The result of these lower downflow velocities may be a higher level of sample cross-contamination using traditional work practice guidelines.
Home » Biosafety Cabinets » [Bulletin] Impact of Class II Biosafety Cabinet Downflow Velocity on Cross Contamination
Shop Biosafety CabinetsView Now
Other content in this Stream
[Video] Dr. Kyle Rohde, Burnett School of Biomedical Sciences, University of Central Florida
In this video interview, Dr. Kyle Rohde, associate professor at the University of Central Florida, discusses his lab's in vitro studies of TB and how they hope to address this global medical crisis.
[Video] Dr. Prakasha Kempaiah, Center for Global Health, Department of Internal Medicine, University of New Mexico
Watch and learn as Dr. Prakasha Kempaiah describes the development of a therapeutic model to assess host genetic factors associated with severe disease outcomes for malaria.
[Video] Marsha Roach, PhoenixSongs Biologicals
Marsha Roach, EVP of PheonixSongs Biologicals, talks about the company’s revolutionary new solution for scientists working with hepatocytes.
[Video] Lizzie Clarke, Center for Global Health, University of New Mexico
Learn about the cell culture methods used by Lizzie Clarke of Dr. Steven Bradfute produce virus-like particles and how these vaccines are tested for their ability to alter immunogenicity.
[Video] Dr. Thomas Loret, University of Manchester
Dr. Thomas Loret of the University of Manchester describes his research into the potential pulmonary toxicity of graphene oxide and other 2D materials, using in vitro methodologies.
[Video] Dr. Aleksandra Dukic, University of Oslo
In this video, hear Dr. Aleksandra Dukic speak on the culture and analysis of glioblastoma patient-derived cells are vital to her work.
[Video] Dr. Theodossis Theodossiou, Institute for Cancer Research, Oslo University Hospital
In this video, Dr. Theodossiou describes how his team are working to improve the efficiency and accuracy of PDT through a variety of methods.
[Drawing] LabGard NU-560/565 Class II Biosafety Cabinet Airflow Schematic
[Drawing] LabGard NU-677-600 Animal Handling Biosafety Cabinet
Product drawing for nominal 6-foot (1.8-meter) width LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-677-600 designed for animal handling use in the vivarium.
[Drawing] LabGard NU-677-500 Animal Handling Biosafety Cabinet
Product drawing for nominal 5-foot (1.5-meter) width LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-677-500 designed for animal handling use in the vivarium.
[Drawing] LabGard NU-677-400 Animal Handling Biosafety Cabinet
Product drawing for LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-677-400 designed for animal handling use in the vivarium.
[Brochure] Animal Laboratory Products
NuAire Lab equipment offers unique, highly dependable, effective, and efficient products for use in the animal vivarium.
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
Product specifications for LabGard Class II Microbiological Safety Cabinet (MBSC) total exhaust model NU-560E (230V).
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
Product specifications for LabGard Class II, Type B2 biosafety cabinet (BSC) total exhaust model NU-560.
[Warranty] Class II, Type B2 Biosafety Cabinet (Worldwide)
Standard warranty information for NuAire Total Exhaust Class II, Type B2 Biosafety Cabinet (BSC) sold outside the United States and Canada.
[Warranty] Class II, Type B2 Biosafety Cabinet (US & CA)
[Spec] LabGard CYTO NU-581E Class II, Type A2 Biosafety Cabinet (230V)
[Spec] LabGard HD ES NU-581 Class II, Type A2 Biosafety Cabinet (115V)
Product specification for LabGard HD (Hazardous Drug) model NU-581 Class II, Type A2 Biosafety Cabinet (BSC).
[Flyer] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet