Over the last several years, Biosafety Cabinet (BSC) manufacturers have lowered downflow velocity while maintaining inflow velocity to increase energy efficiency, reduce exhaust system requirements and lower BSC Noise and vibration levels. However, the lowering of downflow velocity may come with a consequence of greater lateral airflow movement within the work zone creating more significant potential for cross-contamination. The lowering of the downflow velocity does not compromise personnel and product containment performance. Why? Primarily, by maintaining inflow velocity above a certain level as required by the standard, typically 100 fpm (.51 mps). However, the standard does not require a specific downflow velocity and can be adjusted as needed based on the design parameters of the work zone supply diffuser and grill patterns (airflow distribution) to assure compliance to the personnel and product protection requirements of NSF/ANSI 49. NSF/ANSI 49 test procedures do test for cross-contamination, but only at the work zone sidewalls, which doesn’t necessarily translate into the lateral movement of airflow in the center of the work zone. The result of these lower downflow velocities may be a higher level of sample cross-contamination using traditional work practice guidelines.
Shop Biosafety CabinetsView Now
Other content in this Stream
This video will show you how to program the biosafety cabinet lights to turn off automatically after a preset time.
In this product webinar, learn how the LabGard NU-640 Animal Handling Class II, Type A2 Biosafety Cabinet can improve your vivarium facilities workflow while maintaining personnel, product, and enviro
A general rule of thumb is to have your BSC certified annually, after a filter change, when the cabinet is relocated, or the type of work in the cabinet changes.
From the WHO, learn how a lab technican must act in the event of a spill ot power outage when working within a Biosafety Cabinet (BSC).
In this video, Dr. Sofia Forslund, junior group leader at ECRC (MDC/Charité, Berlin), discusses her research into the relationship between the human microbiome and common cardiovascular diseases.
Watch how the airflow in a Class II Biosafety Cabinet (BSC) offers personnel and product protection with the use of smoke.
Watch and learn more about LabGard Biosafety Cabinet model NU-540.
Learn 10 working safely tips when working in a biosafety cabinet.
Product brochure for LabGard ES (Energy Saver) CYTO Class II, Type A2 Biosafety Cabinet model NU-581 featuring a three tired filtration system for use with Cytotoxics, Hazardous Drugs, & API Powders.
The LabGard NU-500 Series Class II Biosafety Cabinets from NuAire provide you the very best in design, ergonomics, and technology.
Minimize workplace injuries working the comfort of a LabGard Biosafety Cabinet.
Work in a safer environment with NuAire electronics monitoring and controlling your work environment.
NuAire Class II Biological Safety Cabinets are designed as a monolithic shell resulting in a long-lasting safe environment for your lab.
Feel at ease working in a longer-lasting safer environment.