NSF/ANSI 49 and BMBL (Biosafety in Microbiological and Biomedical Laboratories) reference the term “Relocated” as the measure of requiring re-certification. Vertical up and down BSC movement on a motorized base stand is not relocation as intended by either of these documents. The height adjustment up and down movement does not cause the vibration necessary to harm or breach the integrity of the HEPA filters. The height adjustment up and down movement is a very slow and smooth movement typical driven by a hydraulic system. If the height adjustment is used in conjunction with an exhausted BSC (i.e. type B1/B2 or canopy connected type A2), the site assessment certification test should include airflow verification throughout the range of motion. The duct connection itself should be made with flexible PVC duct material.
The field certification of Biosafety Cabinets (BSC’s) as required in Annex F of the NSF/ANSI 49 class II bi...
Shop Biosafety CabinetsView Now
Other content in this Stream
LabGard NU-543S Class II, Type A2 Biosafety Cabinet purchase specification.
This video will show you how to program the biosafety cabinet lights to turn off automatically after a preset time.
In this product webinar, learn how the LabGard NU-640 Animal Handling Class II, Type A2 Biosafety Cabinet can improve your vivarium facilities workflow while maintaining personnel, product, and enviro
A general rule of thumb is to have your BSC certified annually, after a filter change, when the cabinet is relocated, or the type of work in the cabinet changes.
From the WHO, learn how a lab technican must act in the event of a spill ot power outage when working within a Biosafety Cabinet (BSC).
In this video, Dr. Sofia Forslund, junior group leader at ECRC (MDC/Charité, Berlin), discusses her research into the relationship between the human microbiome and common cardiovascular diseases.
Watch how the airflow in a Class II Biosafety Cabinet (BSC) offers personnel and product protection with the use of smoke.
Watch and learn more about LabGard Biosafety Cabinet model NU-540.
Learn 10 working safely tips when working in a biosafety cabinet.
Product brochure for LabGard ES (Energy Saver) CYTO Class II, Type A2 Biosafety Cabinet model NU-581 featuring a three tired filtration system for use with Cytotoxics, Hazardous Drugs, & API Powders.
The LabGard NU-500 Series Class II Biosafety Cabinets from NuAire provide you the very best in design, ergonomics, and technology.
Minimize workplace injuries working the comfort of a LabGard Biosafety Cabinet.
Work in a safer environment with NuAire electronics monitoring and controlling your work environment.
NuAire Class II Biological Safety Cabinets are designed as a monolithic shell resulting in a long-lasting safe environment for your lab.