NSF/ANSI 49 and BMBL (Biosafety in Microbiological and Biomedical Laboratories) reference the term “Relocated” as the measure of requiring re-certification. Vertical up and down BSC movement on a motorized base stand is not relocation as intended by either of these documents. The height adjustment up and down movement does not cause the vibration necessary to harm or breach the integrity of the HEPA filters. The height adjustment up and down movement is a very slow and smooth movement typical driven by a hydraulic system. If the height adjustment is used in conjunction with an exhausted BSC (i.e. type B1/B2 or canopy connected type A2), the site assessment certification test should include airflow verification throughout the range of motion. The duct connection itself should be made with flexible PVC duct material.
The field certification of Biosafety Cabinets (BSC’s) as required in Annex F of the NSF/ANSI 49 class II bi...
Other content in this Stream
Read this ebook to learn the Importance of establishing who is responsible for enforcement of lab safety rules, guidelines to keep in mind when conducting a risk assessment, and more.
From the WHO, learn how a lab technican must act in the event of a spill ot power outage when working within a Biosafety Cabinet (BSC).
Learn 10 working safely tips when working in a biosafety cabinet.
Proper preventative maintenance, yearly maintenance, and certification are needed to maximize your products life as well as keep up to date with industry standards to eliminate safety hazards.
After work has been completed proper protocols must be followed to decontaminate the biosafety cabinet to avoid potential contamination risks.
An important part of a safe work environment is making sure it is ergonomically designed to work with you, not against you.
Proper technique when working within the BSC will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
The use of proper procedures in the Biosafety Cabinet (BSC) cannot be overemphasized in providing primary personnel and environmental protection.
Learn steps to take before you begin work in your Biological Safety Cabinet. A proper biosafety or operations manuals will specify practices and procedures designed to minimize and eliminate risks.
Biosafety Cabinets (BSCs) a are designed to remove or minimize exposures to hazardous biological materials and particulates.
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
Learn some of the basic about biosafety cabinets (BSC) and how to go about selecting the right bsc for your application.
Read Strategies to Minimize Contamination in the Cell Culture Lab to learn how to identify potential contamination, steps to control mycoplasma contamination, and more.
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
Product flyer for LabGard AIR Limited 230-volt Class II Biosafety Cabinet (BSC) model NU-545E.
Product flyer for LabGard AIR Limited Class II, Type A2 Biosafety Cabinet (BSC) model NU-545.
Learn the Biosafety Cabinet Decommissioning Process with this infographic.
On June 6, 2016 NSF International published Annex E to Biosafety Cabinetry (BSC) Standard NSF/ANSI 49. Annex E is a free resource for Principal Investigators, Laboratory Managers, Pharmacy...