In today’s world, residential, commercial and institutions (homes, businesses, stores, and schools) commonly install camera surveillance systems for the protection of occupants and visitors. Recently with the news of incidents within pharmacies where the Compounded Sterile Products (CSP’s) were not prepared properly (compromised sterility, incorrect ingredients or concentration) there has been an increased awareness for the use of monitoring or surveillance systems to be located on or near the Primary Engineering Controls (PEC’s) for the purpose of monitoring the very critical work and/or maintaining video evidence on past procedures and training. To accomplish this, NuAire is now offering general guidance to aid in the selection of camera systems that can be mounted on or near the PEC for both new PEC purchases and for field retrofit. In addition to the general guidance, NuAire offers one camera choice along with accessories (i.e. cabling, monitors, DVR’s) for a turnkey solution. Nuaire, as in the past, offers special installation options for camera systems that are provided by the customer. For further information, please contact your NuAire representative, distributor or factory sales team member for assistance.
Use of volatiles in Class II, Type B Biosafety Cabinets (BSCs).
Other content in this Stream
Read this ebook to learn the Importance of establishing who is responsible for enforcement of lab safety rules, guidelines to keep in mind when conducting a risk assessment, and more.
From the WHO, learn how a lab technican must act in the event of a spill ot power outage when working within a Biosafety Cabinet (BSC).
Learn 10 working safely tips when working in a biosafety cabinet.
Proper preventative maintenance, yearly maintenance, and certification are needed to maximize your products life as well as keep up to date with industry standards to eliminate safety hazards.
After work has been completed proper protocols must be followed to decontaminate the biosafety cabinet to avoid potential contamination risks.
An important part of a safe work environment is making sure it is ergonomically designed to work with you, not against you.
Proper technique when working within the BSC will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
The use of proper procedures in the Biosafety Cabinet (BSC) cannot be overemphasized in providing primary personnel and environmental protection.
Learn steps to take before you begin work in your Biological Safety Cabinet. A proper biosafety or operations manuals will specify practices and procedures designed to minimize and eliminate risks.
Biosafety Cabinets (BSCs) a are designed to remove or minimize exposures to hazardous biological materials and particulates.
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
Learn some of the basic about biosafety cabinets (BSC) and how to go about selecting the right bsc for your application.
Read Strategies to Minimize Contamination in the Cell Culture Lab to learn how to identify potential contamination, steps to control mycoplasma contamination, and more.
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
Product flyer for LabGard AIR Limited 230-volt Class II Biosafety Cabinet (BSC) model NU-545E.
Product flyer for LabGard AIR Limited Class II, Type A2 Biosafety Cabinet (BSC) model NU-545.
Learn the Biosafety Cabinet Decommissioning Process with this infographic.
On June 6, 2016 NSF International published Annex E to Biosafety Cabinetry (BSC) Standard NSF/ANSI 49. Annex E is a free resource for Principal Investigators, Laboratory Managers, Pharmacy...