Other content in this Stream
[Spec] NU-481 Specification (115V)
[Spec] NU-540 Class II, Type A2 Biosafety Cabinet (115V)
Product specifications for LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-540 115-volt from NuAire Laboratory Equipment.
[Spec] Class II, Type B2 Biosafety Cabinet NU-430 Specification (115V)
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
[Spec] Class II, Type B1 Biosafety Cabinet Model NU-427 Specification (115V)
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
[Specification] NU-5800 Series Direct Heat CO2 Incubator
Learn the technical side of the NU-5800 Series Direct Heat CO2 Incubator. Dimensions, performance, and more.
[Spec] Class I Containment Ventilated Enclosure NU-813 (115V, 230V)
LabGard model NU-813 is a Class I Containment Ventilated Enclosure providing personnel protection for handling low-risk aerosols and particulate matter.
[White Paper] Centrifuge Maintenance: Protecting the Investment
Centrifuges must be operated correctly and undergo regular maintenance in order to achieve maximum lifespan.
Modify A NuWind Centrifuge Program
Updating or adding a NuWind Centrifuge program can be achieved by selecting the proper icon with your finger located on the touchscreen controls and clicking save.
How To Change A NuWind Centrifuge Rotor Gasket
Gaskets located on a NuWind Centrifuge rotor help to create a secure bond between the lid and the body of the rotor. Replacing a gasket is part of preventative maintenance that will assist as a safety
How To Install A NuWind Centrifuge Bucket Lid
The bucket of a NuWind centrifuge features a cam action type lock to securely create a biosafety seal between the bucket and lid.
How To Change A NuWind Centrifuge Rotor
Changing a rotor on a NuWind Centrifuge can be done in as little as 10 seconds. A twist of a dial, removal of a rotor, placement of a new rotor, an audible "Click" you are done.
How To Install A NuWind Centrifuge Rotor
Removing and installing a rotor on a NuWind centrifuge is as simple as a twist of a dial and listening for an audible "Click".
How To Grease Your Centrifuge
Preventative maintenance such as greasing your centrifuge on a regular basis ensures a long product life and better results.
Initial Startup of a NuWind Centrifuge
Just unpacked your NuWind Centrifuge? Learn the basic setup instructions to power on your new Centrifuge.
[Bulletin] NU-480 Decontamination by Vaporized Hydrogen Peroxide
[Bulletin] NU-480-600E Decontamination Process Testing Results
NU-4950 N2 Gas Consumption
NuAire CO2 Incubator Sterility Test
[Spec] NU-435 Class II, Type B2 Biosafety Fume Hood (115V)