Other content in this Stream
[Spec] NU-481 Specification (115V)
[Spec] NU-540 Class II, Type A2 Biosafety Cabinet (115V)
Product specifications for LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-540 115-volt from NuAire Laboratory Equipment.
[Spec] Class II, Type B2 Biosafety Cabinet NU-430 Specification (115V)
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
[Spec] Class II, Type B1 Biosafety Cabinet Model NU-427 Specification (115V)
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
[Brochure] In-VitroCell CO2 Incubator
In-VitroCell CO2 Incubators from NuAire provide optimal growth conditions for your cells. Available in direct heat and water jacket with popular upgrades such as O2 control, RH Control, and more.
[Bulletin] Clarifying the Class II, Type C1 Biosafety Cabinet
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
[Specification] NU-5800 Series Direct Heat CO2 Incubator
Learn the technical side of the NU-5800 Series Direct Heat CO2 Incubator. Dimensions, performance, and more.
[Spec] Class I Containment Ventilated Enclosure NU-813 (115V, 230V)
LabGard model NU-813 is a Class I Containment Ventilated Enclosure providing personnel protection for handling low-risk aerosols and particulate matter.
[Infographic] Biosafety Cabinet Operating Label
Learn the proper use procedures from start up to shut down when using your Biosafety Cabinet.
[White Paper] Centrifuge Maintenance: Protecting the Investment
Centrifuges must be operated correctly and undergo regular maintenance in order to achieve maximum lifespan.
[Brochure] Hitachi CR22N High Speed Centrifuge
[Brochure] Hitachi FNX Micro Ultracentrifuge
[Brochure] Hitachi CP-WX Ultracentrifuge
[Infographic] 5 Biosafety Cabinet Reminders
We asked NIH National Biosafety and Biocontainment Fellows what some of the most common mistakes made when working with a Biosafety Cabinet.
[Infographic] Hazardous Drug Biosafety Cabinet Operating Procedure
When working in your biosafety cabinet with hazardous drugs it is important to learn the proper procedure from beginning to end.
[Infographic] 10 Tips for Working Safely in the laboratory with Your Biosafety Cabinet
Good technique when working within a Class II Biosafety Cabinet will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
[ebook] 10 Steps to Cleaning a Spill in a Biosafety Cabinet
It happens at some point to even the most seasoned laboratory technician that a spill occurs within the Biosafety Cabinet (BSC). Here are a few steps to cleaning a spill in a BSC.
Modify A NuWind Centrifuge Program
Updating or adding a NuWind Centrifuge program can be achieved by selecting the proper icon with your finger located on the touchscreen controls and clicking save.
How To Change A NuWind Centrifuge Rotor Gasket