Over the last several years, Biosafety Cabinet (BSC) manufacturers have lowered downflow velocity while maintaining inflow velocity to increase energy efficiency, reduce exhaust system requirements and lower BSC Noise and vibration levels. However, the lowering of downflow velocity may come with a consequence of greater lateral airflow movement within the work zone creating more significant potential for cross-contamination. The lowering of the downflow velocity does not compromise personnel and product containment performance. Why? Primarily, by maintaining inflow velocity above a certain level as required by the standard, typically 100 fpm (.51 mps). However, the standard does not require a specific downflow velocity and can be adjusted as needed based on the design parameters of the work zone supply diffuser and grill patterns (airflow distribution) to assure compliance to the personnel and product protection requirements of NSF/ANSI 49. NSF/ANSI 49 test procedures do test for cross-contamination, but only at the work zone sidewalls, which doesn’t necessarily translate into the lateral movement of airflow in the center of the work zone. The result of these lower downflow velocities may be a higher level of sample cross-contamination using traditional work practice guidelines.
Home » Life Science » [Bulletin] Impact of Class II Biosafety Cabinet Downflow Velocity on Cross Contamination
Other content in this Stream
[Video] How to Set Up & Run a Successful Cell Culture Lab
Learn how to set up and run a successful cell culture laboratory by learning topics such as best practices for cell-line validation and considerations for purchasing common cell culture equipment.
[Bulletin] Care and Use of Copper Surfaces
NuAire lab products manufactured with copper such as CO2 Incubators have specific care and use instructions to assure maximum customer satisfaction.
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets and how to go about selecting the right bsc for your application.
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets (BSC) and how to go about selecting the right bsc for your application.
[Spec] NU-677 Class II, Type A2 Animal Handling Biosafety Cabinet Specification (115V)
Product specification for 115-volt LabGard model NU-677Class II, Type A2 Animal Handling Biosafety Cabinet.
[Spec] NU-481E Specification (230V)
[Spec] NU-481 Specification (115V)
[Spec] NU-540E Class II Microbiological Safety Cabinet (230V)
[Spec] NU-540 Class II, Type A2 Biosafety Cabinet (115V)
[Spec] NU-543E Class II Microbiological Safety Cabinet (230V)
[Spec] Class II, Type B2 Biosafety Cabinet NU-430 Specification (115V)
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
[Spec] Class II, Type B1 Biosafety Cabinet Model NU-427 Specification (115V)
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
[White Paper] Use of HEPA Filters in Biosafety Cabinets
Learn about the use of HEPA filters in biosafety cabinets.
[Flyer] Mini Microcentrifuges
Mini Microcentrifuges from NuAire Laboratory Equipment are available in three different models all at varying maximum speeds.
[Flyer] NuAire Research Laboratory Equipment Line Card
NuAire Laboratory Equipment is the safer choice for your lab. View NuAire's core line of research products with this line card.
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
[Spec] NU-PR797 Specification (115v)
[Brochure] In-VitroCell CO2 Incubator
In-VitroCell CO2 Incubators from NuAire provide optimal growth conditions for your cells. Available in direct heat and water jacket with popular upgrades such as O2 control, RH Control, and more.
[Bulletin] Clarifying the Class II, Type C1 Biosafety Cabinet
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
[Specification] NU-5800 Series Direct Heat CO2 Incubator