Over the last several years, Biosafety Cabinet (BSC) manufacturers have lowered downflow velocity while maintaining inflow velocity to increase energy efficiency, reduce exhaust system requirements and lower BSC Noise and vibration levels. However, the lowering of downflow velocity may come with a consequence of greater lateral airflow movement within the work zone creating more significant potential for cross-contamination. The lowering of the downflow velocity does not compromise personnel and product containment performance. Why? Primarily, by maintaining inflow velocity above a certain level as required by the standard, typically 100 fpm (.51 mps). However, the standard does not require a specific downflow velocity and can be adjusted as needed based on the design parameters of the work zone supply diffuser and grill patterns (airflow distribution) to assure compliance to the personnel and product protection requirements of NSF/ANSI 49. NSF/ANSI 49 test procedures do test for cross-contamination, but only at the work zone sidewalls, which doesn’t necessarily translate into the lateral movement of airflow in the center of the work zone. The result of these lower downflow velocities may be a higher level of sample cross-contamination using traditional work practice guidelines.
Shop Pharmacy EquipmentClick Here
Other content in this Stream
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
Learn the proper use procedures from start up to shut down when using your Biosafety Cabinet.
Watch what items you should check off your list once your pharmacy isolator arrives at your loading dock.
We asked NIH National Biosafety and Biocontainment Fellows what some of the most common mistakes made when working with a Biosafety Cabinet.
When working in your biosafety cabinet with hazardous drugs it is important to learn the proper procedure from beginning to end.
Good technique when working within a Class II Biosafety Cabinet will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
It happens at some point to even the most seasoned laboratory technician that a spill occurs within the Biosafety Cabinet (BSC). Here are a few steps to cleaning a spill in a BSC.
The focus of this article is to discuss the details for consideration of a good working design of a USP 800 compliant compounding lab.
Changing gloves on your Compounding Aseptic Containment Isolator must be done on a routine basis. Follow these simple steps to safely change your CACI gloves.
Airflow in a negative pressure compounding aseptic containment isolator (CACI) is designed to keep you safe.
Testing the adjustable base stand of your new pharmacy isolator is a great way to ensure your product is workly correctly.
Maintain sterility in your PharmaGard isolator by avoiding disrupting air over items located on the work surface.
Work in a safer environment with NuAire electronics monitoring and controlling your work environment.
Minimize workplace injuries working the comfort of a LabGard Biosafety Cabinet.
LabGard Biosafety Cabinets (BSCs) are constructed to last longer keeping you and your laboratory safe.
The LabGard NU-500 Series Class II Biosafety Cabinets from NuAire provide you the very best in design, ergonomics, and technology.
NuAire offers general guidance to aid in the selection of camera systems that can be mounted on or near your primary engineering controls.