The In-VitroCell CO2 Incubator line from NuAire are offered in Direct Heat and Water Jacketed temperature control models. Sizes range from 7 cubic feet (200 Liters) to 5.65 cubic feet (160 Liters). In-VitroCell CO2Incubators offer the quickest recovery times on the market and provide the reliability needed so you can concentrate on what matters; your research.
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and ...
Other content in this Stream
Product specifications for LabGard Class II, Type A2 Biosafety Cabinet (BSC) model NU-540 115-volt from NuAire Laboratory Equipment.
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
Learn the technical side of the NU-5800 Series Direct Heat CO2 Incubator. Dimensions, performance, and more.
LabGard model NU-813 is a Class I Containment Ventilated Enclosure providing personnel protection for handling low-risk aerosols and particulate matter.
Learn the proper use procedures from start up to shut down when using your Biosafety Cabinet.
Centrifuges must be operated correctly and undergo regular maintenance in order to achieve maximum lifespan.
We asked NIH National Biosafety and Biocontainment Fellows what some of the most common mistakes made when working with a Biosafety Cabinet.
When working in your biosafety cabinet with hazardous drugs it is important to learn the proper procedure from beginning to end.
Good technique when working within a Class II Biosafety Cabinet will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
It happens at some point to even the most seasoned laboratory technician that a spill occurs within the Biosafety Cabinet (BSC). Here are a few steps to cleaning a spill in a BSC.
Updating or adding a NuWind Centrifuge program can be achieved by selecting the proper icon with your finger located on the touchscreen controls and clicking save.
Gaskets located on a NuWind Centrifuge rotor help to create a secure bond between the lid and the body of the rotor. Replacing a gasket is part of preventative maintenance that will assist as a safety
The bucket of a NuWind centrifuge features a cam action type lock to securely create a biosafety seal between the bucket and lid.