The TankGuard™ automatic tank switch has been designed to provide reliable service to perform the critical backup gas (i.e. CO2, N2) tank switch for a tissue cell culture incubator. The TankGuard™ tank switch continuously monitors and indicates the selected source gas tank and the status of the non-selected or secondary source gas tank. If the selected source gas tank pressure falls below 8 PSI, the selected source internal pressure switch will detect the low pressure and automatically change sources.
Most Recent Flipbooks
[Instructions] Stainless Steel Water Pan NU-5842 Operation and Installation
Installation and operation instructions for NuAire CO2 Incubator stainless steel water pan model number NU-5842.
[Infographic] Telescoping Base Stand Assembly
Learn how to properly assemble a telescoping base stand from NuAire to fit anyone of our Biosafety Cabinets (BSC), Laminar Airflow Workstations (LAFW), and more.
[Instructions] NU-1553 CO2 Incubator Stacking Table Operation and Maintenance Manual
The NU-1553 stacking table is designed to support a second Autoflow or In-VitroCell CO2 Incubator over the top of the first one when purchased separately.
[Instructions] CO2 Incubator Copper Water Pan NU-5860 Instructions
Learn how to use and install a CO2 Incubator copper water pan from NuAire Lab Equipment.
[Instructions] NU-1582 Mobile Platform for CO2 Incubators