It happens at some point to even the most seasoned laboratory technician that a spill occurs within the Biosafety Cabinet (BSC). Taking precautionary measures before and during your work with hazardous materials could help keep you and others safe. Remember, if a spill occurs, don’t panic. Here are some simple steps to keep you and your laboratory safe. Please check with your Environmental Health and Safety (EHS) office or Biosafety Officer to ensure you have proper steps in place in case of a spill based of standard Biosafety in Microbiological and Biomedical Laboratories (BMBL).
Home » Working Safely in Your Biosafety Cabinet » [ebook] 10 Steps to Cleaning a Spill in a Biosafety Cabinet
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in M...
Shop Biosafety CabinetsView Now
Other content in this Stream
[Webinar] Biosafety Cabinet Introduction Webinar
Learn some of the basic about biosafety cabinets (BSC) and how to go about selecting the right bsc for your application.
[Video] Chapter 1: Introduction to Biosafety Cabinets
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
[Video] Chapter 2: Understanding the Biosafety Cabinet
Biosafety Cabinets (BSCs) a are designed to remove or minimize exposures to hazardous biological materials and particulates.
[Video] Chapter 3: Proper Preparation - The 4 Steps to Take Before You Begin Work in Your Biosafety Cabinet
Learn steps to take before you begin work in your Biological Safety Cabinet. A proper biosafety or operations manuals will specify practices and procedures designed to minimize and eliminate risks.
[Video] Chapter 4: Using Proper Procedures and Techniques with the Biosafety Cabinet
The use of proper procedures in the Biosafety Cabinet (BSC) cannot be overemphasized in providing primary personnel and environmental protection.
[Video] Chapter 5: Working in your Biosafety Cabinet
Proper technique when working within the BSC will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
[Video] Chapter 5a: Biosafety Cabinet Ergonomics
An important part of a safe work environment is making sure it is ergonomically designed to work with you, not against you.
[Video] Chapter 6: When Work is Completed in the Biosafety Cabinet
After work has been completed proper protocols must be followed to decontaminate the biosafety cabinet to avoid potential contamination risks.
[Video] Chapter 7: Maintenance and Certification of a Biosafety Cabinet
Proper preventative maintenance, yearly maintenance, and certification are needed to maximize your products life as well as keep up to date with industry standards to eliminate safety hazards.
[Video] BenchTip: 10 Tips to Working Safely in your Biosafety Cabinet
Learn 10 working safely tips when working in a biosafety cabinet.
[Infographic] 10 Tips for Working Safely in the laboratory with Your Biosafety Cabinet
Good technique when working within a Class II Biosafety Cabinet will minimize air turbulence and prevent splatter or unwanted spread of aerosols.
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in Microbiological and Biomedical Laboratories), a clear 12-inches (305 mm) should be present...
[Tip] Why You Should Work at the Proper Biosafety Cabinet Sash Window Level
Class II Biological Safety Cabinets, or BSCs, provide vital personal, product, and environmental protections in the laboratory setting. BSCs go through a certification process to meet the...
[Presentation] Tips for Working Safely in your Biosafety Cabinet
Learn a few tips about how to work safely in your Biosafety Cabinet (BSC).
[Tip] Never Cover the Airflow Grill on a Biosafety Cabinet
The front airflow grill on a biological safety cabinet performs the important job of separating the clean air inside the cabinet from the contaminated laboratory air outside.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Infographic] Biosafety Cabinet Operating Label
Learn the proper use procedures from start up to shut down when using your Biosafety Cabinet.
[Infographic] 5 Biosafety Cabinet Reminders
We asked NIH National Biosafety and Biocontainment Fellows what some of the most common mistakes made when working with a Biosafety Cabinet.
[Bulletin] When to Replace My Biosafety Cabinet