Setting the UV Light Timer on the Aeromax Control System
The video on how to set up and use the Ultraviolet (UV) Light timer feature on the Aeromax control system of a NuAire Biosafety Cabinet shows a laboratory technician or lab manager how to program the UV light to turn off automatically after a preset time. First, the user is instructed to enter the UV timer menu using a specific key sequence, and a blinking light will indicate that the user has successfully entered the menu. Then, using arrow keys, the user can increase or decrease the time the UV light should remain on and then press the wave key to accept the time or the silence key to abort. Once the UV light is turned on, it will automatically turn off after the preset time elapses, and a blinking light above the UV key will indicate that the timer is active.